Firing of chopping gap by alternative methods.
AbstractsFiring of electrodes at impulse generators or chopping gap systems is usually carried out by the external electrode ignition system. The electrode systems are optimally set at a greater mutual distance than the spontaneous flashover. And the flashover of impulse occurs only when the space between the electrodes is ionized by external device. This method is very reliable, inexpensive and easily adjustable at single stage electrode systems. However, in the testing laboratories are often used multistage generators and electrode systems for the possibility of work with a higher peak value of impulse voltage. Firing of electrodes at these systems is complicated by the standard external electrode ignition system. Thus it may be better to use one of the alternative methods, preferably in wireless design. The main function of an alternative system is to create an ionized medium between system electrodes. It is possible to use UV germicidal lamps, UV lasers, powerful lasers or microwave energy and ionized radiation.
File Size: 678,5 KB
may download free of charge all publications including most recent ones
may download free of charge publications over three years old and purchase all publications
Challenges in the control center (EMS) due to distributed generation and renewables
Control methodologies for direct voltage and power flow in a meshed HVDC grid
Framework for EPU operators to manage the response to a cyber-initiated threat to their critical infrastructure
MO surge arresters - Metal oxide resistors andsurge arresters for emerging system conditions
Ground potential rise at overhead AC transmission line structures during power frequency faults
Voltage Source Converter (VSC) HVDC for Power Transmission - Economic Aspects and Comparison with other AC and DC Technologies
Update of Service Experience of HV Underground and Submarine Cable Systems
Integration of large Scale Wind Generation using HVDC and Power Electronics
Recommendations for testing of long AC submarine cables with extruded insulation for system voltage above 30 (36) to 500 (550) kV (This TB replaces electra article ELT_189_1)
For any questions in connection with the on-line library, contact:
Publications and Editions Manager